Bruker Ultraflex TOF/TOF, year 2001/2002
Currently non operational due to defective slave processor and lift pulser.
Recently refurbished with NEW LTB MNL 200 laser and NEW PIE box, NEW rough pump. Before slave processor failure worked fine in linear and reflector modes
Vacuum in lower 10 x e-7 range – turbo pumps and rough pumps in excellent condition.
NEW parts installed in May 2017:
- LTB MNL 200 laser
- PIE II pulser
Extra spare parts:
- Ion suurce, mounted
- Laser attenuator, mounted
- Valve set, sourc/analyzer/rough
- Loading mechanism with motors
PC with digitizer board (1 channel only working) and FC 3.0 SW
Lift not operational due to defective lift pulser II.
Sells as a whole unit or for parts.
System is in Northern Ireland, EU.